Air pressure transmitter can accurately measure atmospheric pressure value, suitable for pressure measurement under various circumstances. Adopt imported sensors, measurement data is stable, high precision, strong anti-interference ability, long service life and can be widely applied to computer room environment, airports, railway stations, commercial buildings, family homes, office buildings, schools, meeting rooms, shopping malls, restaurants, the gym, cinema, library, etc, it has broad application prospects.
* Model: JXBS-3001- QY-V10–1
* Minimum order quantity: 1 piece
* Delivery time: within 24 hours
1.Designed for real-time monitoring of ambient temperature, humidity and air pressure.
2.Wall-mounted, easy to use.
3.Provides Modbus RS485 communication ports.
4.12-24V DC power supply.
Measurement parameter: atmospheric pressure
Model Number: JXBS-3001-QY-V10-1
Communication mode: 0-10V
Power supply: 12-24V power supply
Measurement range: 10-1200mbar
Maximum power: <4W
Accuracy: + / – 1.5 mbar
Working range: 0-40℃/0-80RH%
Power consumption: 0.15 W or less
Place the sensor in an environment that is sheltered from wind and rain, and hang it at 90 degrees perpendicular to the floor. Keep the air hole of the sensor facing directly below to prevent water intake.
Wall-mounted prince shell is wall-mounted, the mounting hole is located in the middle of both sides of the equipment, the mounting aperture is less than 4mm, the hole distance is 105mm, can use 3mm self-tapping screw installation.
Air pressure sensor cannot be connected to PLC or computer
1, the computer has more than one COM port, select the wrong port.
2, air pressure sensor address is not correct, or there is a duplicate device address (all factory default is 1).
3, baud rate, check mode, data bit, stop bit error.
4. The host polling interval and waiting response time are too short. Set them to more than 200ms.
5. The 485 bus is disconnected, or the A and B wires are connected in reverse.
6, if the number of air pressure sensor is too much or the connection is too long, should be the nearest power supply, add 485 intensifier, increase the resistance of 120Ω end.
7. The USB to 485 driver is not installed or damaged.
8. Equipment damage.